Dzerzhynsk Instrument-Manufacturing Plant
- Address:Business Center, DNU, 24 Universytetska str., c. Donetsk, 83000, Ukraine
- Telephone:+38(062) 381-03-50, 381-03-51
- manufacture, installation and service of electromagnetic meters of electroconducting fluids designed to be mounted on the pipes with a diameter range of 80-4,000 mm
The flowmeter is an up-to-date device for volume measurement and the measurement of volume consumption of the electroconducting fluids (water, mixture, pulp, ets.), running in the pipes with a diameter of 80 mm and over, within the temperature range of -2 C - +130 C. This is a small-sized and easy to install device (the installation of the device is 10-25 times as cheap as that of fluid flowmeters in operation). The flowmeter consists of the reorganizer ISP-204, which measures the velocity of fluid stream and transforms it into a unified signal (0.5 to 5 mA), Ц the given reorganizer is recorded in the State Measuring Instruments List, item number U1190-90, - and the analogue concordance block BS-1 that is a secondary measuring-reorganizer. The storage and generalization of the exploitation process, the analysis of the impressions, proposals and wishes of the consumers enabled us to produce devices meeting the present-day standards. Taking into account demands of consumers, up-to-date modifications of the Block BC were designed, which are compatible with IBM PC/AT computers both as a separate block on the basis of PC or as a part of PC.
The consumption-counter works in the following operation conditions given in the table 1:
Operation conditions Block ISP Block BS
Air temperature , C -10 to +45 -10 to +35
The temperature of medium measured, C -2 to +130
Atmospheric pressure, kPa 84 to 106 84 to 106
Length of the chain ISB-concordance block
BS with a resistance in the mains of up
to 20 Om, in the measured mains of up
to 200 Om, m 200
Relative humidity, % <=100 <=75
At temperature, C <=30 <=35
Velocity range measured, m/s 0.015 Е 2.2
Inner diameter of the pipe, mm 80 Е 4000
Main relative error of the velocity measurement, % 1.0
Range of changes in the out-going signal, mA 0 Е 5.0
Power consumption <= 0.13
Range of electroconductivity of mediums
measured, Om/m 10-3 Е 10
By now over 300 devices have been installed at the Ukrainian plants. All these devices have been highly evaluated. Among the plants using the flowmeters are:
Ц JSC УObolonФ - 3 devices have been installed to keep the record of beer consumption and that of water in the artesian wells.
Ц Druzhkovka Machine-Building Plant Ц 3 devices have been installed to keep the record of water consumption (For 6 years they have been working without a break-down).
Ц OJSC УCrimea Soda-Making PlantФ Ц on the pulp, containing 12% of hard components and up to 30% of salt within the temperature range of +8 Е +90 C.
Ц УMakeyevska UkrpromvodchermetФ (they have been working for more than 5 years)
омпани€ "ј–ћј“ј" существует 12 лет. «а это врем€ мы многого достигли и прошли путь от небольшой мастерской до современного производства, оснащенного европейским оборудованием. Ќаш опыт и технологии позвол€ют производить разнообразную, ƒ≈…—“¬»“≈Ћ№Ќќ высококачественную продукцию.
- LoginS Trading Company
- Lauffer Group Company
- Patriot Air Company
- Medtorg Scientific Production Enterprise
- POLETEHNIKA Company
- BHV Analytical Company
- Advocates law firm
- Metropoliya Group Companies
- Grace-Engineering Company
- Ppromarmatura Company
- SUMY-ELECTRODE LTD
- DEL POST Ltd
- Inkotel Group Company
- Atomcoplect Separated Subdivision NNEGC "Energoatom"
- Pan Ќоtеl
- Psyol Hotel
- V.P. Komisarenko Institute of Endocrinology and Metabolism of the National Academy of Medical Sciences of Ukraine
- Cooperator Hotel
- HyperHost company
- Lars Export Company